Model 1007C Temperature Chamber Specifications
|Cool Down Transition Time*|
|Start Temp||End Temp|
|+23°C||-----||4 min||18 min||25 min||33 min||Ultimate|
|+50°C||5 min||11 min||27 min||36 min||45 min||Ultimate|
|+85°C||13 min||20 min||37 min||47 min||55 min||Ultimate|
|+150°C||32 min||39 min||58 min||65 min||76 min||Ultimate|
|Heat Up Transition Time*|
|Start Temp||End Temp|
|+23°C||-----||1.5 min||7 min||14 min||20 min||25 min|
|0°C||1.5 min||3.5 min||13 min||20 min||23 min||31 min|
|-40°C||6 min||11 min||17 min||24 min||30 min||35 min|
|-55°C||8 min||13 min||19 min||26 min||32 min||37 min|
|-65°C||10 min||14 min||21 min||28 min||34 min||39 min|
Rate Of Change
To calculate rate of change for a particular condition, take the difference between the Start Temp and End Temp and divide by the Transition Time.
Cool Down Example: From +50°C to -40°C = 90°C / 27 min = 3.33°C/min.
Heat Up Example: From -40°C to +50°C = 90°C / 11 min = 8.18°C/min.
*Note: Transition times are measured after a 2 hour soak at the respective start temperature with an empty chamber, as indicated on the temperature controller, 23°C ambient. Measured with setpoint beyond the start and end temperatures. Does not include the effect of proportional band when approaching setpoint. Performance is reduced by 17% with 50 Hz input power.
|Live Load Capacity|
|1,000 Watts||800 Watts||500 Watts||400 Watts||300 Watts|
|Refrigeration and Heating System|
|High Stage Refrigerant||R-404A (Dupont HP-62)|
|Low Stage Refrigerant||R-508B (Dupont SUVA-95)|
|Compressors||1.5 HP x 1.5 HP Tecumseh hermetic compressors in a cascade configuration.|
|Heat of Rejection||14,800 BTUH (maximum rated chamber load at maximum cooling rate from high temperature soak)|
|Heater Power||2,000 Watts|
|Air Flow||450 cfm|
|Temperature Controller||Watlow F4T Touch Screen Controller with RS-232, Ethernet interface, 4.3" color graphic touch screen. OR...
Watlow F4 Controller with RS-232 interface, LED readout of temperature, LCD display of other parameters.
|Limit Controller||Independent high and low temperature limits. Triggers an audible alarm and shuts down the chamber. Relay contacts provide a safety power interlock for test sample.|
|Chart Recorder||(Optional) Honeywell DR4300 Series. One pen, 10" circular chart. Mounts in lower front door.|
|Input Power Requirements|
|Phase||Current Draw||Minimum Service|
|230 V ±10%, 60 Hz||1 PH||25 A||30 A|
|3 PH||19 A||25 A|
|208 V -5/+10%, 60 Hz||1 PH||28 A||35 A|
|3 PH||21 A||30 A|
Input may be configured for single or three phase in the field by changing jumpers. Three phase load is semi-balanced. Call for 50 Hz operation. Performance is reduced by 17% with 50 Hz input power.
Customer power source must be hard-wired to the chamber by a qualified electrician. Power cord and plug is not included.
|Physical Characteristics and Safety|
|Inside Dimensions||24" W x 21" H x 24" D (7 cubic feet)
609 mm W x 533 mm H x 609 mm D (198 liters)
|Outside Dimensions*||33" W x 67.75" H x 54.25" D (nominal)
838 mm W x 1721 mm H x 1378 mm D
|* Door latch adds 3" to width on right side (may be removed to permit move-in through a 36" doorway). Circulator motor and housing adds 6" to height.|
|Minimum Installed Clearance||18" from the left and right side
24" from the rear
|Window Viewing Area||13.375" W x 9" H|
|Access Ports||4" Port on left and right side (two total)
Supplied with foam plugs
|Weight||Chamber Weight: 850 pounds
Shipping Weight: 1,011 pounds
|Sound Level||62 dBA in cooling mode (A-weighted, measured 36" from the front or side surface, 63" from the floor, in a free-standing environment)|
NOTE: Performance is typical and based on operation at 23°C (73°F) ambient and nominal input voltage. Designed for use in a normal conditioned laboratory. Operation at higher ambient temperatures may result in decreased cooling performance. Additional ports and shelves will also affect performance. Operation above 30°C (85°F) or below 16°C (60°F) ambient is not recommended.