Model 123H Temperature/Humidity Chamber - Specifications
Standard Range: 10% to 95%
(Limited by a 6°C dewpoint and maximum dry bulb of +85°C)
With GN2 Purge or optional Dry Air Purge: 5% to 95%
(Dry bulb range of +10°C to +85°C)
NOTE: Ability to reach RH extremes may be limited by the humidity sensor accuracy. Low Dew Point conditions can only be achieved when starting with a clean, dry chamber.
Cool Down Transition Time* (uncontrolled humidity mode)
|Start Temp||End Temp|
|+23°C||-----||6 min||31 min||51 min||68 min||Ultimate|
|+50°C||7 min||16 min||44 min||62 min||82 min||Ultimate|
|+85°C||18 min||28 min||54 min||82 min||101 min||Ultimate|
|+150°C||43 min||54 min||87 min||108 min||129 min||Ultimate|
|Heat Up Transition Time* (uncontrolled humidity mode)|
|Start Temp||End Temp|
|+23°C||-----||2 min||12 min||24 min||33 min||Ultimate|
|0°C||3 min||9 min||18 min||32 min||41 min||Ultimate|
|-40°C||9 min||16 min||25 min||39 min||48 min||Ultimate|
|-55°C||14 min||18 min||31 min||44 min||53 min||Ultimate|
|-65°C||15 min||21 min||32 min||45 min||54 min||Ultimate|
Rate Of Change
To calculate rate of change for a particular condition, take the difference between the Start Temp and End Temp and divide by the Transition Time.
Cool Down Example: From +85°C to -40°C = 125°C / 54 min = 2.3°C/min.
Heat Up Example: From -40°C to +85°C = 125°C / 25 min = 5.0°C/min.
*Note: Transition times are measured after a 2 hour soak at the respective start temperature with an empty chamber, as indicated on the temperature controller, 23°C ambient. Measured with setpoint beyond the start and end temperatures. Does not include the effect of proportional band when approaching setpoint. Measured at nominal line voltage of 120V, 60 Hz. Performance is reduced on 123H-EX 50 Hz export versions by 17%.
Live Load Capacity (uncontrolled humidity mode)
|500 Watts||400 Watts||225 Watts||175 Watts||100 Watts|
Refrigeration and Heating System
|High Stage Refrigerant||R-404A (Dupont HP-62)|
|Low Stage Refrigerant||R-508B (Dupont SUVA-95)|
|Compressors||1/2 HP x 1/3 HP Tecumseh hermetic compressors in a cascade configuration.|
|Heat of Rejection||5,000 BTUH (maximum rated chamber load at maximum cooling rate from high temperature soak)|
|Air Heater Power||1,000 Watts (500 Watts when compressors are running)|
|Humidifier Heater Power||750 Watts|
|Temp/Humidity Controller||Watlow F4T Touch Screen Controller with RS-232, Ethernet interface, 4.3" color graphic touch screen. OR...
Watlow F4 Controller with RS-232 interface, LED readout of temperature, LCD display of other parameters.
|Limit Controller||Independent high and low temperature limits. Triggers an audible alarm and shuts down the chamber. Relay contacts provide a safety power interlock for test sample.|
|Chart Recorder||(Optional) Honeywell DR4300 Series. Two pen, 10" circular chart. Mounts in lower front door. 24-hour rotation.|
Input Power Requirements
123H North America Version:
123H-EX Export Version:
6' Power cord supplied with a molded NEMA 5-20P plug.
Plugs into a standard NEMA 5-20R receptacle.
Model 123H-EX export version is supplied with power cord for the destination country.
Humidity Water Requirements
|Supply and Drain||Must be provided with a water line and floor drain. Negligible consumption.|
|Water Recirculation System (optional)||Provides a reliable supply of filtered water for the humidity system. Perfect for installations where a water line and drain are not available.|
|Inside Dimensions||18" W x 16.5" H x 13.5" D (2.3 cubic feet)|
|Outside Dimensions||26" W x 63" H x 36.5 " D (nominal)
Vent tube adds 3" to height.
|Minimum Installed Clearance||6 " from the left and right side
12" from the rear
|Window Viewing Area||7" W x 12 " H|
|Access Ports||4" Port on left and right side (two total)
Supplied with foam plugs
|Weight||Chamber Weight: 520 pounds
Shipping Weight: 635 pounds (North America only, contact us for export shipment)
|Sound Level||62 dBA (A-weighted, measured 36" from the front)|
NOTE: Performance is typical and based on operation at 23°C (73°F) ambient and nominal input voltage. Designed for use in a normal conditioned laboratory. Operation at higher ambient temperatures may result in decreased cooling performance. Additional ports and shelves will also affect performance. Operation above 30°C (85°F) or below 16°C (60°F) ambient is not recommended.
Due to continuous product development, specifications are subject to change without notice.